
Parameters for C:\MassLynx\DESI.PRO\ACQUDB\20260309_021_M2L_9-LHS_pos04 on XevoTQAbsolute DESI MRM.EXP

Data Processing:
SIR / MRM Chromatogram Spike Removal	ON
SIR / MRM Smoothing		OFF
Smoothing window size (scans)	3
Number of smooths		2


Prescan Statistics:
Initial Average Intensity	3681.9008
Initial Average Std Dev		0.7854
Bunch Zero Level		0.0231
Bunch Std Dev			0.9221
Bunch Threshold			7.3769
Spike Removal Std Dev		0.0000


Method Events:
Initial Stop Flow:	No Change
Initial Switch 2:	No Change
Initial Switch 3:	No Change
Initial Switch 4:	No Change
Initial Infusion:	No Change
Initial Flow State:	LC
Initial Flow Rate:	5
Initial Reservoir:	No Action
API Probe Delay Temp:	20
Initial Refill:		No Action
Timed Events Disabled
Event	Time(Sec)	Name		Action


Source Information:
Type - DESIXSHTL
Status - Option Connector Disconnected

Ion Count Threshold Positive:	150.0000
Ion Count Threshold Negative:	300.0000
Ion Count Threshold (adjusted for gain):	190.0000


[DESI Experiment Parameters]
DesiXStart		-10.9987
DesiYStart		-1.7645
DesiXLength		6.7461
DesiXStep		0.0050
DesiXRate		245.0000
DesiYLength		3.0274
DesiYStep		0.0350
DesiSlot		A

Instrument Parameters - Function 1:
Parameter File - C:\MassLynx\DESI.PRO\ACQUDB\DESI_general.IPR
Polarity	ES+
Calibration	Static 2
Soft Transmission Mode Disabled
Capillary (kV)	0.70	0.81
Cone (V)	25.00	107.30
Source Offset (V)	30.0
Source Temperature (°C)	150	150
HTL Temperature (°C)	125	125
Cone Gas Flow (L/Hr)	NA
Purge Gas Flow (L/Hr)	NA
NanoFlow Gas Flow (Bar)	NA
Nebuliser Gas Flow (L/Hr)	ON
LM 1 Resolution	2.8
HM 1 Resolution	14.9
Ion Energy 1	0.4
MS Mode Collision Energy	4.00
MSMS Mode Collision Energy	15.00
MS Mode Entrance	1.00
MS Mode Exit	1.00
Gas On MS Mode Entrance	1.00
Gas On MS Mode Exit	1.00
Gas On MSMS Mode Entrance	1.00
Gas On MSMS Mode Exit	1.00
Gas Off MS Mode Entrance	30.00
Gas Off MS Mode Exit	30.00
Gas Off MSMS Mode Entrance	30.00
Gas Off MSMS Mode Exit	30.00
LM 2 Resolution	2.7
HM 2 Resolution	15.1
Ion Energy 2	0.5
Gain	1.00
Multiplier	585.59
Active Reservoir	Wash
Cone Voltage Ramp :Disabled
Probe Temperature Ramp :Disabled


Engineers Settings:
MS1 Low Mass Position      	672
MS1 High Mass Position     	281
MS1 Low Mass Resolution    	218
MS1 High Mass Resolution   	1582
MS1 Resolution Linearity   	528
MS1 High Mass DC Balance   	0.20
MS1 DC Polarity            	Positive
MS2 Low Mass Position      	671
MS2 High Mass Position     	290
MS2 Low Mass Resolution    	220
MS2 High Mass Resolution   	1655
MS2 Resolution Linearity   	528
MS2 High Mass DC Balance   	0.20
MS2 DC Polarity            	Negative


Inter-scan delays:
Automatic Mode
MS Inter-scan delay (secs)                   Automatic
Polarity/Mode switch Inter-scan delay (secs) Automatic
Enhanced Inter-scan delay (secs)             Automatic
Inter-channel delay - See Tables
MS 1 Delay Table:
         R     delay
<=   1.250     0.001
<=   4.000     0.002
<=  10.000     0.003
<=  20.000     0.004
>   20.000     0.005
